Calibration and Adjustment
As a manufacturer, we have the necessary test equipment and the expertise to calibrate and calibrate the magnetic field meters and probes we manufacture.
If we find out during the calibration that an adjustment is necessary, we will do it. The adjustment made is also noted on the factory calibration certificate.
We deliver to our instruments a factory calibration certificate without measured values as a calibration certificate according to ISO 9001: example of a calibration certificate for one of our AS-active probes.
We offer for our teslameters and magnetic field probes the calibration with traceability to the national standards of the Physikalisch-Technische Bundesanstalt (PTB ). Traceability is confirmed by our calibration certificate.
We also specify the measurement uncertainty of the calibration performed in our factory calibration certificate.
Our test equipment monitoring is subject to the ISO 9001-certification.
Option Linearity Curve
As an optional additional service, we offer the creation of a linearity curve for our AS active probes. In doing so, we record the individual fault course of the probe as a function of the strength of the magnetic field. With many probes, the linearity curve is also created at different temperatures.
Calibration Interval for our Teslameters and Magnetic Field Probes
For the calibration interval, no general statement can be made. The interval is u.a. depending on the frequency of use and the resulting stress. A laboratory device used only twice a year must certainly be calibrated less frequently than a device used every day in production.
The requirements of the own test equipment management the definition of the calibration interval.
By default we recommend an annual calibration for our products.